class resistics.time.reader_spam.TimeReaderSPAM(dataPath: str)[source]

SPAM data has the following characteristics:

• SPAM raw data is single precision floats with unit Volts.

• Getting unscaled samples returns data with unit mV for both the electric and magnetic fields. This is because gain is removed in unscaled samples to ensure consistency when a single recording is made up of multiple data files, each with different gain settings

• The start time in XTR files is the time of the first sample in the data

• The end time in XTR files is the time of the last sample in the data

In situations where a SPAM dataset is recorded in multiple small files, it is required that the recording is continuous.

Notes

Getting unscaled samples for SPAM data removes the gain rather than return exactly the values in the data files. In cases where there are multiple data files, it is not necessary that they have been recorded with the same gain. Therefore, to ensure consistency when looking at raw data, the gain is removed at the getUnscaledSamples stage rather than getPhysicalSamples, where it would have probably been more appropriate. This means that getUnscaledSamples returns data where all channels are in mV.

The scalings to convert the raw data to mV are stored in the ts_lsb chan header and calculated out as the header files are being read.

Todo

Attributes
recChannelsDict

Channels in each data file

dtypenp.float32

The data type

numDataFilesint

The number of data files

Methods

chanDefaults(self) → Dict[str, Any][source]

Returns
Dict[str, Any]

Dictionary of headers for channels and default values

getDataFilesForSamples(self, startSample: int, endSample: int) → Tuple[List[str], List[List[int]], List[float]][source]

Get the data files that have to be read to cover the sample range

Parameters
startSampleint

Starting sample of the sample range

endSamplesint

Ending sample of the sample range

Returns

Time data object

getPhysicalSamples(self, **kwargs)[source]

Get data scaled to physical values

resistics uses field units, meaning physical samples will return the following:

• Electrical channels in mV/km

• Magnetic channels in mV

• To get magnetic fields in nT, calibration needs to be performed

Parameters
chansList[str]

List of channels to return if not all are required

startSampleint

First sample to return

endSampleint

Last sample to return

remaveragebool

Remove average from the data

remzerosbool

Remove zeroes from the data

remnans: bool

Remove NanNs from the data

Returns
TimeData

Time data object

Notes

The method getUnscaledSamples multiplies the raw data by the ts_lsb converting it to mV. Because gain is removed when getting the unscaledSamples and all channel data is in mV, the only calculation that has to be done is to divide by the dipole lengths (east-west spacing and north-south spacing).

To get magnetic fields in nT, they have to be calibrated.

getUnscaledSamples(self, **kwargs) → resistics.time.data.TimeData[source]

Get raw data from data file, returned in mV

SPAM raw data is single precision float with unit Volts. Calling this applies the ts_lsb calculated when the headers are read. This is because when a recording consists of multiple data files, each channel of each data file might have a different scaling. The only way to make the data consistent is to apply the ts_lsb scaling.

Therefore, this method returns the data in mV for all channels.

Parameters
chansList[str], optional

List of channels to return if not all are required

startSampleint, optional

First sample to return

endSampleint, optional

Last sample to return

Returns
TimeData

Time data object

headersFromRawFile(self, rawFile: str, headers: Dict) → None[source]

Read the headers from the raw file and figure out the data byte offset.

Parameters
rawFilestr

The .RAW data file

Notes

Open with encoding ISO-8859-1 because it has a value for all bytes unlike other encoding. In particular, want to find number of samples and the size of the header. The extended header is ignored.

mergeHeaders(self, headersList: List, chanHeadersList: List) → None[source]

Checks all the header files to see if there are any gaps and calculates the sample ranges for each file together with the total number of samples. Sets the start and end time of the recording and class variables datetimeStart and datetimeStop.

Parameters

List of headers from each data file

List of chan headers from each data file

printDataFileInfo(self) → None[source]

Print a list of the data files

printDataFileList(self) → List[str][source]

Information about the data files as a list of strings

Returns
List[str]

List of information about the data files

readHeader(self) → None[source]

For SPAM data, the may be more than one header file as data can be split up into smaller files as it is recorded. In that case, the header information should be somehow merged.

All sampling frequencies should be the same

readHeaderXTR(self, headerFile: str) → None[source]

The raw data for SPAM is in single precision Volts. However, if there are multiple data files for a single recording, each one may have a different gain. Therefore, a scaling has to be calculated for each data file and channel. This scaling will convert all channels to mV.

For the most part, this method only reads recording information. However, it does additionally calculate out the lsb scaling and store it in the ts_lsb channel header. More information is provided in the notes.

Parameters

Notes

The raw data for SPAM is in single precision floats and record the raw Voltage measurements of the sensors. However, if there are multiple data files for a single continuous recording, each one may have a different gain. Therefore, a scaling has to be calculated for each data file.

For electric channels, the scaling begins with the scaling provided in the header file in the DATA section. This incorporates any gain occuring in the device. This scaling is further amended by a conversion to mV and polarity reversal,

$scaling = read scaling from DATA section of header file \ scaling = 1000 * scaling , \ scaling = -1000 * scaling , \ ts_lsb = scaling ,$

where the reason for the 1000 factor in line 2 is not clear, nor is the polarity reversal. However, this information was provided by people more familiar with the data format.

For magnetic channels, the scaling in the header file DATA section is ignored. This is because it includes a static gain correction, which would be duplicated at the calibration stage. Therefore, this is not included at this point.

$scaling = -1000 , \ ts_lsb = scaling ,$

This scaling converts the magnetic data from V to mV.

readHeaderXTRX(self, headerFile)[source]

Parameters

setParameters(self) → None[source]

spamHeaders(self) → Tuple[List[str], Dict[str, str]][source]